HeartSine Samaritan PAD AED, Pads, Batteries & Accessories...click
HeartSine Samaritan PAD PAK-01
Authorized Heartsine Direct Distributor
Medical Devices, Inc. Company
The Most Trusted Name In AED Defibrillator Supplies
30 Day Unconditional
Shop with confidence
We Accept Purchase Order
Wall Signs |
Samaritan Package Deals
Replacement New Adult Heartsine Samaritan Pad-Pak cartridge:
Heartsine samaritan PAD-PAK
Combined Electrode and Battery Cartridge integrated unit
is gray in color and is intended to be used with
suspected victims of SCA for patients age 8 years to adult and weighing >25 kg/55 lbs.
Only one expiry date to monitor. Replacement of the battery
and defibrillation pack must be carried out if the expiry
date of the Pad-Pak™ has been exceeded. If the status
indicator on the samaritan® PAD is not flashing you may need
to replace the Pad-Pak™. For diagnosis of the reason for
the status indicator not flashing, please refer to the
troubleshooting section of your owner’s manual.
Adult Electrodes (Pad-Pak-01)
HeartSine samaritan PAD-PAK adult cartridge comes complete with
battery and electrode pads as one integrated
Cartridge typically has a shelf-life of 3.5
years from date of manufacture, offering
significant savings over other
defibrillators that require separate battery
and pad units.
WHEN TO REPLACE THE PAD-PAK™:
The battery lifetime in use can be up to 6
hours monitoring or 30 shocks or a
combination of both. A Pad-Pak™ in the stand
by mode (inserted into the samaritan® PAD)
has a shelf-life indicated by the expiration
date (typically years from manufacture).
Replacement < of the battery and
defibrillation pack must be carried out if:
the expiry date of the Pad-Pak > has been
exceeded the Pad-Pak has been used ( is a
single use item) it must be replaced with a
new Pad-Pak ™ cartridge. If the status
indicator on the samaritan® PAD is not
flashing you may need to replace the Pad-Pak
Fo. r diagnosis of the reason for the status
indicator not flashing please refer to the
troubleshooting section of this manual.
Maintenance Replacing The Pad-Pak The
battery lifetime in use can be up to 6 hours
monitoring, 60 shocks or a combination of
both. A Pad-Pak in stand by mode (inserted
into the SAM 300P) has a shelf-life
indicated by the expiry date (4 years from
manufacture date)*. The Pad-Pak must be
replaced out if: If the status indicator on
the SAM 300P is flashing red or not flashing
you may need to replace the Pad-Pak. For
diagnosis of the reason for the status
indicator flashing red or not flashing
please refer to the troubleshooting section
of this manual.
HOW TO REPLACE A PAD-PAK™
1.Take the replacement Pad-Pak™ from its
protective bag. 2.Remove the old Pad-Pak™ which is to be
3.Follow the instruction for Pad-Pak™
installation which can be
found on the page titled “Preparing your
samaritan® PAD for use”.
4. Push the Pad-Pak™ firmly to ensure it is
5.Check status indicator. If the Pad-Pak™
has been inserted
correctly, status indicator flashes green
approximately every 5
6.Press the On/Off button to turn the device
on. Listen for the
appropriate messages to start. Press the
On/Off button again to
turn the device off. Ensure no warning
messages are issued by the
device and that the status indicator
continues to flash
once every five seconds.
7. If necessary inform relevant safety
officer or person responsible
for maintenance of the samaritan® PAD.
8. Update the relevant records to show the
date that the replacement
Pad-Pak™ was placed into service.
9.Dispose of the old Pad-Pak™.
PAD-PAK™ OR PEDIATRIC-PAK™.
HeartSine Technologies have developed two
versions of the Pad-
Pak™. The standard Pad-Pak™ is intended to
be used with
suspected victims of SCA who are over eight
years old or weigh more
than 25kg (55lb).
The Pediatric-Pak™ (child) is intended for
us on suspected victims of
SCA who are over one year old and less than
eight years old weighing
less than 25kg (55lb). The Pediatric-Pak™ is
designed for use only in
the samaritan® PAD SAM300P.
HeartSine recommend that a samaritan® PAD
sam300P is deployed
in locations where emergency responses
involving pediatric patients
may occur. Contact your authorized
distributor or HeartSine directly
for further information.
You must remember to check the expiration date on the
The Pad-Pak™ is a single use device with a
stated expiry date. The
expiry date is stated on the Pad-Pak™. The
expiry date is given beside
the symbol shown in the figure on the left
Pad-Pak™ has a warranty for use up until the
date stated. The Pad- Pak™ should not be
used beyond the stated expiry date. The
Pad-Pak™ is a single use item. If a
Pad-Pak™ is used on a patient
they must not be used again.
Check the expiration date of the Pad-Pak™
inserted into the samaritan® PAD. If the
has exceeded its use by date, remove it from
samaritan® PAD and replace with a new Pad-Pak™.
Remove Pad-Pak™ from its packaging.
Push Pad-Pak™ into the opening and
listen for the “click” sound to ensure it
is properly inserted.
Place the samaritan®
PAD into its Soft Carry Case.
For alternative transport cases ask
CHECK DEVICE is working optimally.
Button. Ensure you can
hear the voice prompts:
CALL for medical assistance
by pressing the Button
PAD-PAK™ INSTALLATION INSTALL 4
Put into a wall case or safe visible
The Pad-Pak™ includes the battery and
defibrillation electrodes in
cartridge. See expiry date for both on
Place the samaritan® PAD and the Pad-Pak™™
on a flat surface.
Once the Pad-Pak™ is installed properly
the PAD STATUS INDICATOR will begin to
blink Green every 5 seconds.
Do not open Pad-Pak™ tray or open
protective packaging until the time of
when they are applied to a patient.
HeartSine Samaritan Pad Pediatric® Pad-Pak™
Integrated battery & electrode System
Better Business Bureau Accredited
Of Commerce, Palatine, Illinois, U.S.A.
Member U.S. Woman's Chamber Of Commerce
348 W. Colfax St., Palatine IL 60067
Phone: 1-847-202-3858 Fax:1-866-879-7795. Toll
Copyright © 2004 - 2011 All rights of design and text