1141-000147-non-rechargeable replacement Battery Pak (alternate battery pak number: 11141-000158) is specifically designed for use with the LIFEPAK 500 AED. Includes instructions for removal of the LIFEPAK 500 battery, installation and safe disposal of the Battery Pak. Battery discharger is also included.
The LIFEPAK 500 AED can be powered by two types of batteries:
A rechargeable sealed lead-acid battery or one of two non-rechargeable lithium batteries (sulfur dioxide or manganese dioxide) provide power to the AED. The rechargeable battery requires periodic recharging by an external battery charger.
LIFEPAK 500 non-rechargeable sealed lead-acid (SLA) battery pak 11141-000147 or (11141-000158) .
Note: Either type of battery may be installed. Follow the guidelines described in Physio-Control LIFEPAK 500 owner's manual to help maximize battery life and performance. Use only Medtronic Battery Pak batteries with the LIFEPAK 500 AED. And please, do not use beyond the expiration date marked on the battery label. Most LIFEPAK 500 AEDs include a readiness display on the device's handle that can be seen at all times. OK displays if the automatic self-test is completed successfully. If the self-test detects that service is required or if the device detects that the battery needs immediate replacement, the OK indicator disappears and a service and/or battery indicator appear(s).
WARNING! Please note that the LIFEPAK 500 non-rechargeable lithium manganese dioxide battery pak does not fit in all LIFEPAK 500 AEDs. Use only with AEDs marked -003 inside the battery well. When the LIFEPAK 500 AED prompts REPLACE BATTERY, replace the battery immediately. Using an improperly maintained battery to power the AED may cause power failure without warning. Maintain batteries always.