Physio-Control LIFEPAK 15 defibrillator Lithium-ion Battery - 5.7 Ah 5.7 amp hour, 11.1 volt, rechargeable with fuel gauge. For use with the Physio-Control LIFEPAK 15 monitor/defibrillator. Lifepak-15 STATION BATTERY CHARGER is for use with Lithium-ion batteries for the LIFEPAK 15 monitor/defibrillator. This innovative and highly automated systems provide rapid charging and ease of use. Charging automatically begins when the battery is inserted. There are no buttons to push. Quiet and convenient Whisper quiet operation does not distract from patient care. A mounting bracket is included for space-saving installation. STATION BATTERY CHARGER is supplied with 110-240 V AC power cord for use in stationary applications Alternate Part Number: 3206735-000
Physio-Control LIFEPAK 15 Monitor-Defibrillator Lithium Ion Rechargeable Fuel Gauge Battery 21330-001176: The LIFEPAK 15 monitor/defibrillator operates either on battery power using two Lithium-ion batteries, or with auxiliary power using the AC Power Adapter or DC Power Adapter. Batteries may be charged in the Station or Mobile Li-ion Battery Charger, the REDI-CHARGE(TM) Battery Charger, or in the monitor/defibrillator if it is connected to auxiliary power.
Note: Although the monitor/defibrillator can operate using auxiliary power with no batteries installed, at least one battery should be installed at all times. If the monitor/defibrillator loses power for more than 30 seconds, the device reverts to the user-configured default settings and begins a new patient record. IMPORTANT! The LIFEPAK 15 monitor/defibrillator Lithium-ion batteries are not interchangeable with batteries that are used in other LIFEPAK defibrillators. Routinely inspect batteries for damage or leakage. Recycle or discard damaged or leaking batteries. Each battery has a fuel gauge that indicates the approximate charge level in the battery. Press the gray button above the battery symbol to check the battery's charge level prior to installing it in the defibrillator. The four battery indicators shown here represent approximate charge--greater than 70%, greater than 50%, greater than 25%, and 25% or less, respectively. POSSIBLE DEVICE SHUTDOWN may occur if the device battery is not fully charged, properly maintained. Always have immediate access to a spare, fully charged, properly maintained battery. Replace the battery when the device displays a low battery warning. Please observe the battery charging indicator. LED is illuminated when installed batteries are fully charged. LED flashes when either battery is charging. LED is not illuminated when no batteries are installed or a battery is unable to be charged.
Note: Older or heavily used batteries lose charge capacity. If Physio-Control LIFEPAK 15 Monitor-Defibrillator Lithium Ion Rechargeable Fuel Gauge Battery 21330-001176 indicates fewer than four LEDs immediately after completing a charge cycle, the battery has reduced capacity. If the battery fuel gauge shows two or fewer LEDs after the battery completes a charge cycle, the battery should be replaced.